# Capacitance measurement

## Capacitance measurement

The measurement of the ratio of the charge induced on a conductor to the change in potential with respect to a neighboring conductor which induces the charge. In a multiconductor system there are capacitances between each pair of conductors. In general, these capacitances are functions of the total geometry, that is, the location of all of the conducting and dielectric bodies. When, as is usually true, only the capacitance between two conductors is of interest, the presence of other conductors is an undesirable complication. It is then customary to distinguish between two-terminal and three-terminal capacitors and capacitance measurements. In a two-terminal capacitor, either one of the conductors of primary interest surrounds the other (in which case the capacitance between them is independent of the location of other bodies except in the vicinity of the terminals); or the somewhat indefinite contributions of the other conductors to the capacitance of interest are accepted.

A three-terminal capacitor consists of two active electrodes surrounded by a third, or shield, conductor. The direct capacitance between the two active electrodes is the capacitance of interest, and, when shielded leads are used, it is independent of the location of all other conductors except the shield.

Every physically realizable capacitor has associated loss in the dielectric and in the metal electrodes. At a single frequency these are indistinguishable, and the capacitor may be represented by either a parallel or series combination of pure capacitance and pure resistance. The measurement of capacitance, then, in general involves the simultaneous measurement of, or allowance for, an associated resistive element. See Permittivity

Most capacitance measurements involve simply a comparison of the capacitor to be measured with a capacitor of known value. Methods which permit comparison of essentially equal capacitors by simple substitution of one for the other at the same point in a circuit are frequently possible and almost always preferable.

#### Bridge comparison methods

When capacitors must be compared with high accuracy, bridge methods must be adopted. See Bridge circuit, Wheatstone bridge

Resistance-ratio bridges are Wheatstone-bridge configurations in which the potential division of the capacitor being measured and either a parallel combination of a standard loss-free capacitor Cs and a conductance Gs or a series combination of Cs and a resistor Rs is equated, when the detector is nulled, to the ratio of potentials across resistors R1 and R2. More commonly now, the reference potential division is that of a variable-ratio autotransformer known as an inductive voltage divider (IVD). See Inductive voltage divider

The Schering bridge yields a measurement of the equivalent series-circuit representation of a capacitor.

The resistance-ratio and Schering bridges are useful for two-terminal capacitance measurements. Their use may be extended to three-terminal measurements and extended in accuracy and range by the introduction of shielding and the addition of the Wagner branch.

#### Time-constant methods

If a direct voltage is suddenly applied to the series combination of a resistor and an initially discharged capacitor, the charge and the voltage on the capacitor increase exponentially toward their full magnitudes with a time constant equal in seconds to the product of the resistance in ohms and the capacitance in farads. Similarly, when a charged capacitor is discharged through a resistor, the charge and the voltage decay with the same time constant. Various methods are available for the measurement of capacitance by measurement of the time constant of charge or discharge through a known resistor. See Time constant

In one such method the time required for the output voltage of an operational amplifier having a capacitor as a feedback component to increase to a value equal to the step-function input voltage applied through a resistor to its input is determined by an electronic voltage-comparison circuit and timer. With the assumption of ideal characteristics for the amplifier, such as infinite gain without feedback, infinite input impedance, and zero output impedance, the measured time interval is equal to the product of the values of the known resistance and the capacitance being measured.

Mentioned in ?
References in periodicals archive ?
999, Resistance Measurement 20 Ohm To 20 Mega Ohm, Capacitance Measurement 200 Pf 2000 Uf, Induction Measurement 200 Uh To 200H With One Pair Test Leads And Battery Installed Similar To Kusam Percent U2013 Meco Model 954 Mk Percent U2013 1 Along With User Manual, Warranty And Calibration Certificate
Reliable touch detection provides extremely high sensitivity with very low capacitance measurement resolution, offering a highly accurate and robust solution.
It is based on capacitance measurement of the water molecules in the resin.
For organic coatings, the capacitance measurement is particularly revealing.
During the filling stage, most of the melt has a temperature greater than the melt temperature, so the influence of temperature on the capacitance output is small, and a near linear relation between the capacitance measurement and the melt position can be assumed.
This paper presents the results of a study into the detection of transverse moisture gradients in timber using a method of precision capacitance measurement at a frequency of 1 MHz.
The uncompensated acetate group has a strong dipole moment and is detectable in a capacitance measurement.
The Agilent scanning microwave microscopeand field emission scanning electron microscope incorporate a powerful atomic force microscope and a PNA to provide accurate scanning capacitance measurement at nanoscale.
Limited Tenders are invited for Repair Of Doble Make Tan Delta And Capacitance Measurement Test
They deliver accurate measurement of current, voltage, and frequency on all electrical systems as well as capacitance measurement for start and run motor capacitors, and resistance and continuity to verify the quality of electrical connections and check motor and transformer coils.

Site: Follow: Share:
Open / Close