# circuit testing

## circuit testing

[′sər·kət ‚tes·tiŋ]
(electricity)
The testing of electric circuits to determine and locate an open circuit, or a short circuit or leakage.

## Circuit testing (electricity)

The testing of electric circuits to determine and locate any of the following circuit conditions: (1) an open circuit, (2) a short circuit with another conductor in the same circuit, (3) a ground, which is a short circuit between a conductor and ground, (4) leakage (a high-resistance path across a portion of the circuit, to another circuit, or to ground), and (5) a cross (a short circuit or leakage between conductors of different circuits). Circuit testing for complex systems often requires extensive automatic checkout gear to determine the faults defined above as well as many quantities other than resistance.

In cable testing, the first step in fault location is to identify the faulty conductor and type of fault. This is done with a continuity tester, such as a battery and flashlight bulb or buzzer (Fig. 1), or an ohmmeter.

Simple continuity test setup

Useful for locating faults in relatively low-resistance circuits, the Murray loop is shown in Fig. 2 with a ground fault in the circuit under test. A known “good” conductor is joined to the faulty conductor at a convenient point beyond the fault but at a known distance form the test connection. One terminal of the test battery is grounded. The resulting Wheatstone bridge is then balanced by adjusting RB until a null is obtained, as indicated by the detector in Fig. 2. Ration RA/RB is then known. For a circuit having a uniform ratio of resistance with lenght, circuit resistance is directly proportional to circuit lenght. Therefore, the distance to the fault is determined from the producer given by Eqs. (1)–(3). From Eq. (3) and a knowledge of total length l of the circuit, once ratio r has been measured, the location of the fault x is determined. (1)  (2)  (3)

Murray loop for location of ground fault

The Varley loop test is similar to the Murray loop test except for the inclusion of the adjustable resistance R. The Varley loop is used for fault location in high-resistance circuits.

An alternating-current capacitance bridge can be used for locating an open circuit as shown in Fig. 3. One test terminal is connected to the open conductor and the other terminal to a conductor of known continuity in the cable. All conductors associated with the test are opened at a convenient point beyond the fault but at a known distance from the test connection. An audio oscillator supplies the voltage to the bridge, which is balanced by adjusting RB for a null as detected by the earphones. Measured ratio RA/RB equals the ratio of capacitances between the lines and the grounded sheath. Because each capacitance is proportional to the length of line connected to the bridge, the location of the open circuit can be determined from Eq. (4). (4)  See Circuit (electricity)

Alternating-current capacitance bridge used in location of an open circuit in one conductor
Mentioned in ?
References in periodicals archive ?
VEIKI-VNL has around 60 employees and is a member of the Short-Circuit Testing Liaison, the association of companies for short circuit testing laboratories.
short circuit testing, high voltage testing, temperature rise testing, as well as mechanical and aging tests.
NIST has provided fabrication development advice and circuit designs and circuit testing.
The cost estimate for bare board manufacture is based on complexity and includes costs for internal and external circuit generation, lamination, AOI testing, hole drilling, solder masking, legend printing, and final circuit testing.
a leading testing service provider that focuses on integrated circuit testing for wireless and communication applications, has purchased multiple Fusion CX systems.
CopperMax, working hand-in-hand with Spirent's REACT(R) OSS, is used by many of the worlds leading telephone companies for end-to-end DSL circuit testing.
As part of the development, Alcatel is providing an interface to its 7300/7301 Advanced Services Access Manager (ASAM) Digital Subscriber Line access multiplexers (DSLAMs), while Spirent is producing test hardware that conducts the actual circuit testing.
The NIL demonstrates real-world service management processes, including customer relationship management, order management and provisioning, circuit testing and billing for managed IP broadband services.
First fill of reagents is underway and "wet" circuit testing will begin shortly.
As a part of this agreement, an LTX Fusion HF will be installed at ISE Labs, an ASE Test company and Silicon Valley's largest integrated circuit testing laboratory.
Cell test methods include, but are not limited to charging, discharging, capacity, storage performance evaluation, cycle life, reliability tests, short circuit testing and overcharge/discharge testing.

Site: Follow: Share:
Open / Close