For example, the stagnation temperature uncertainty component is calculated by
The LP CFV stagnation temperature in Stage 2 is measured using a Rosemount 162N100 A resistance temperature device (RTD).
The MP CFV stagnation temperature in Stage 2 is also measured using a Rosemount 162N100A resistance temperature device (RTD).
In this work, a heat exchanger is used to control the Stage 2 stagnation temperature to within 2 K of its Stage 1 value.
Measurements of the Nozzle Bank Stagnation Temperature and TMS Temperature
The uncertainty of the nozzle bank stagnation temperature has an additional uncertainty component attributed to the correction in calculating [T.
Uncertainty of the Stage 1 stagnation temperature YSI Model 46000 Thermistor Relative uncertainty Absolute Percent LP CFV stagnation temp.