# Temperature measurement

Also found in: Wikipedia.

## Temperature measurement

Measurement of the hotness of a body relative to a standard scale. The fundamental scale of temperature is the thermodynamic scale, which can be derived from any equation expressing the second law of thermodynamics. Efforts to approximate the thermodynamic scale as closely as possible depend on relating measurements of temperature-dependent physical properties of systems to thermodynamic relations expressed by statistical thermodynamic equations, thus in general linking temperature to the average kinetic energy of the measured system. Temperature-measuring devices, thermometers, are systems with properties that change with temperature in a simple, predictable, reproducible manner. See Temperature, Thermodynamic principles

In the establishment of a useful standard scale, assigned temperature values of thermodynamic equilibrium fixed points are agreed upon by an international body (General Conference of Weights and Measures), which updates the scale about once every 20 years. Thermometers for interpolating between fixed points and methods for realizing the fixed points are prescribed, providing a scheme for calibrating thermometers used in science and industry.

The scale now in use is the International Temperature Scale of 1990 (ITS-90). Its unit is the kelvin, K, arbitrarily defined as 1/273.16 of the thermodynamic temperature T of the triple point of water (where liquid, solid, and vapor coexist). For temperatures above 273.15 K, it is common to use International Celsius Temperatures, t90 (rather than International Kelvin Temperatures, T90), having the unit degree Celsius, with symbol °C. The degree Celsius has the same magnitude as the kelvin. Temperatures, t90, are defined as t90/°C = T90/K - 273.15, that is, as differences from the ice-point temperature at 273.15 K. The ice point is the state in which the liquid and solid phases of water coexist at a pressure of 1 atm (101,325 pascals). [The Fahrenheit scale, with symbol °F, still in common use in the United States, is given by tF/°F = (t90/°C × 1.8) + 32, or tF/°F = (T90/K × 1.8) - 459.67.] The ITS-90 is defined by 17 fixed points.

Primary thermometers are devices which relate the thermodynamic temperature to statistical mechanical formulation. The fixed points of ITS-90 are all based on one or more types of gas thermometry or on spectral radiation pyrometry referenced to gas thermometry. Secondary thermometers are used as reference standards in the laboratory because primary thermometers are often too cumbersome. It is necessary to establish standard secondary thermometers referenced to one or more fixed points for interpolation between fixed points. Lower-order thermometers are used for most practical purposes and, when high accuracy is required, can usually be calibrated against reference standards maintained at laboratories, such as the U.S. National Institute of Standards and Technology, or against portable reference devices (sealed boiling or melting point cells). See Gas thermometry, Low-temperature thermometry, Pyrometer, Thermistor, Thermocouple, Thermometer

McGraw-Hill Concise Encyclopedia of Physics. © 2002 by The McGraw-Hill Companies, Inc.
References in periodicals archive ?
'Temperature measurement helps optimize processes and detect failures, reducing downtime and improving operational efficiency,' said NETL's Jessica Mullen, who managed the project.
Summary: The patient temperature measurement market has been analyzed in detail to offer definitive forecast insights on the market for the period 2017-22.
A literature review of CINAHL, Cochrane Database of Systematic Reviews, and PubMed was completed using the keywords temperature measurement or device, temperature, fever, and thermometers.
It enables the possibility of setting up individual customized temperature measurement ranges.
Non-contact temperature measurement: The new device shows its strength above all with exact temperature measurement of particularly hot objects or objects that are difficult to access, such as in the steel, plastics or glass industries.
[10] showed that temperature measurement by WMS technology can be applied under high pressure.
Thermistor temperature sensors have high output signal, high precision, and low requirements for the secondary instrument, but their disadvantage is a small useful temperature measurement range [8, 9].
(EDL) is the worldwide leader in high quality temperature measurement devices and calibrators.
Variohm EuroSensor's ERTD line of platinum thin film RTD sensors are available in a choice of PT100, PT500 and PT1000 elements for precision temperature measurement across a range of -70[degrees]C to +500[degrees]C continuous, with tolerance class options of 2B, B, A, 1/3B and 1/10B.
Precise temperature measurement and control is important in many fields of the science and industry, including biochemical, material sciences, energy accounting, process control.
A solar powered digital industrial thermometer has been added to the company's line of industrial temperature measurement solutions.

Site: Follow: Share:
Open / Close