theodolite(redirected from Transit (surveying))
Also found in: Dictionary, Thesaurus.
theodolite(thēŏd`əlīt'), calibrated optical instrument used to determine relative position in surveyingsurveying,
method of determining accurately points and lines of direction (bearings) on the earth's surface and preparing from them maps or plans. Boundaries, areas, elevations, construction lines, and geographical or artificial features are determined by the measurement of
..... Click the link for more information. , navigation, and meteorology. It is similar in construction to a surveyor's transit, consisting of a telescope fitted with a spirit level and mounted on a tripod so that it is free to rotate about its vertical and horizontal axes. (Sometimes two or more telescopes of different magnifications are used.) Graduated scales are used to measure the amount of rotation about the axes. Measurements of the altitude and azimuth of a weather balloonweather balloon,
balloon used in the measurement and evaluation of mostly upper atmospheric conditions (see atmosphere). Information may be gathered during the vertical ascent of the balloon through the atmosphere or during its motions once it has reached a predetermined maximum
..... Click the link for more information. at precise time intervals are used to compute the estimated wind velocity of the atmosphere.
(or transit), a geodetic instrument for determining directions and measuring horizontal and vertical angles during geodetic work and topographic and mine surveying, in construction, and in other applications. Horizontal and vertical circles calibrated in degrees and smaller units are the main measuring devices in the theodolite.
Until the mid-20th century, theodolites with metal circles read by means of verniers or micrometric microscopes were used. In the 1920’s, theodolites with glass circles equipped with optical reading devices appeared, and they came to be called optical theodolites. Schematic and optical diagrams of a theodolite are given in Figures 1 and 2, respectively; the devices for the vertical circle analogous to those for the horizontal circle are not shown.
In the USSR, GOST (the All-Union State Standard) permits only the manufacture of optical theodolites, whose main specifications are given in Table 1 (the number included in the type name is the permissible mean square error of measurement of the horizontal angle in seconds of arc).
Theodolites often have various attachments, such as a declinometer, sighting marks, or an optical range finder.
|Table 1. Specifications of main Soviet theodolites|
|Diameter of circles (mm)||Scale divisions||Magnification of telescope||Maximum measurement of Vertical angles||Weight in case (kg)|
|Note: The reading devices of the T05, T1, and T2 are optical micrometers, that of the T5 and T15 are scale microscopes, and that of the T30 is an index|
|T05 ...............||180||130||10’||1”||35 ×||50°||21 + 15|
|T1 ...............||135||90||10’||1”||30 ×||65°||13.5|
|T2 ...............||90||65||20’||1”||25 ×||75°||95|
|T5 ...............||95||70||1°||1°||28 ×||65°||6 5|
|T15 ...............||72||72||1°||2’||25 ×||60°||40|
|T30 ...............||72||72||10’||–||20 ×||55°||3.2|
Various kinds of specialized theodolites exist. Among them are astronomical theodolites, which allow sighting on the zenith and have ocular micrometers; tachymeters, which automatically give the difference in elevation of points according to readings on a scale; mine surveying theodolites, for work in mine shafts; gyroscopic theodolites, for determining the direction of the meridian; and theodolites that automatically record the results on punched tape for feeding to a computer.
The theodolite has a number of typical instrument errors, whose effect is diminished by well-thought-out design, careful manufacture and testing, and appropriate measurement techniques.
REFERENCESGOST 10529-70: Teodolity. Tipy. Osnovnye parametry i teknicheskie trebovaniia.
GOST 20063-74: Teodolity. Metody ispytanii i proverki.
Eliseev, S. V. Geodezicheskie instrumenty ipribory, 3rd ed. Moscow, 1973.
Deimlich, F. Geodezicheskoe instrumentovedenie. Moscow, 1970. (Translated from German.)
Zakharov, A. I. Novye teodolity i opticheskie dal’nomery. Moscow, 1970.
G. G. GORDON