# noise figure

Also found in: Acronyms, Wikipedia.

## noise figure

(noise factor) See noise.

## Noise Figure

(or noise factor), the numerical characteristic of a radio receiver that indicates how much its sensitivity to an incoming signal is decreased by the effects of its internal noise (seeFLUCTUATIONS, ELECTRICAL).

The noise figure F is equal to the ratio of the total noise power at the output of an actual receiver Pact to the output noise power of a similar ideal (noiseless) receiver Pid under the condition that the only source of input noise in both cases is the thermal noise of the matched impedance (dummy antenna) at the temperature T0 = 290°K (seeNYQUIST FORMULA):

(1) F = Pact/Pid = Pact/kT0 Δ fG

Here, k is Boltzmann’s constant, Δf is the receiver’s passband in hertz, and G is the receiver’s power gain. The noise figure is also expressed in decibels: F (dB) = 10 log F. For an ideal receiver, F = 1 (or 0 dB), while for an actual receiver, F > 1. Often, the noise figure is described by the noise temperature Tn.

The noise figure is measured by means of standard noise generators or standard signal generators by determining how much the total power of the receiver’s output signal is increased with a calibrated signal Pgen at its input compared with the output power without the signal. When measuring by the “twofold excess” method, Pgen is adjusted so that the output-signal power is doubled. Then, Pact = GPgen, and the noise figure is calculated from equation (1).

Quantum mechanical amplifiers and cooled parametric amplifiers using semiconductor diodes, for which F ≈ 1.1 and Tn ≈ 30°K (when not cooled, F ≈ 1.3 and Tn ≈ 100°K), have the lowest noise figures. For amplifiers using traveling-wave tubes and tunnel diodes, F ≈ 3 to 10 and Tn ≈ 600° to 3000°K. Radio and television receivers have noise figures ranging from several units to several tens.

### REFERENCES

Kuz’min, A. D. Izmerenie koeffilsienla shuma priemno-usilitel’nykh ustroistv. Moscow-Leningrad, 1955.
Sukhodoev, I. V. Shumy eleklricheskikh tsepei (Teoriia). Moscow, 1975.

I. T. TROFIMENKO

## noise figure

[′nȯiz ‚fig·yər]
(electronics)
References in periodicals archive ?
First two low noise amplifiers (LNA) are low power/low noise, pHEMTS MMICs internally matched to 50[ohm] having noise figure of 2.
Using a combination of a high-end R&S ZVA and the R&S ZVAX-TRM, users can now measure the noise figures of amplifiers, converters and transmit/receive (T/R) modules for radar and satellite applications.
The most common method used for measuring noise figure is the Y-factor approach.
Noise figure measurements enable test engineers to determine potential degradation of signal-to-noise ratio from system components in applications including radar, wireless communications, and digital communications.
In addition to the balanced amplifier application, these devices are a perfect solution for the first and second stages of a base station LNA due to the excellent combination of low noise figure and linearity.
Noise figure as a measurement parameter came into use in the early 1940s.
The noise figure of a receiver or an amplifier is specified by its manufacturer, but the determination of the system's noise figure is a little more complicated.
Agilent's high-performance, network-analyzer-based noise figure measurement technique is the most accurate available today," said Gregg Peters, vice president of Agilent's Component Test Division.
This allows designers to make tradeoffs between operating current and output linearity while maintaining an optimum noise figure.
The cost of the good LO-IF isolation of the Gilbert-cell type mixer is a higher noise figure than for a single balanced mixer for the same power consumption.
MwT), a wholly owned subsidiary of IXYS Corporation (NASDAQ:IXYS), announced that it has introduced a family of three AlGaAs/InGaAs based low noise pHEMT devices with extremely low noise figures with operational frequency up to 38 GHz.
The three-way APS3608 exhibits a typical noise figure of 5 dB with gain of 5.

Site: Follow: Share:
Open / Close