# ohmmeter

Also found in: Dictionary, Thesaurus, Medical, Wikipedia.

## ohmmeter

ohmmeter (ōmˈmēˌtər), instrument used to measure the electrical resistance of a conductor. It is usually included in a single package with a voltmeter, and often an ammeter. In normal usage, the ohmmeter operates by using the voltmeter to measure a voltage drop, then converting this reading into a corresponding resistance reading through Ohm's law. If the current is known, the voltage drop across the unknown resistance may be read to give the resistance directly. If the current is not known, or if it is not possible to measure the voltage across the unknown directly, reading the voltage drop across a known resistance in the same circuit will give the current, and once this voltage is subtracted from the total drop for the circuit, the voltage drop across the unknown, and thus its resistance, may be found.

## Ohmmeter

A portable instrument for measuring relatively low values of electrical resistance. The range of resistance measured is typically from 0.1 microhm to 1999 ohms (&OHgr;). The ohmmeter solves quickly and easily a variety of measurement problems, including measuring the resistance of cladding and tracks on printed circuit boards, electrical connectors, and switch and relay contacts, as well as determining the quality of ground-conductor continuity and bonding, cables, bus-bar joints, and welded connector tags. See Resistance measurement

McGraw-Hill Concise Encyclopedia of Physics. © 2002 by The McGraw-Hill Companies, Inc.
The following article is from The Great Soviet Encyclopedia (1979). It might be outdated or ideologically biased.

## Ohmmeter

a direct-reading instrument for measuring electric true (ohmic) resistances. Among the types of ohmmeter are megohmmeters, teraohmmeters, and microhmmeters, which differ in the ranges of the resistances being measured. Ohmmeters are made with a permanent-magnet measuring head or a permanent-magnet quotient meter.

The operation of a permanent-magnet ohmmeter is based on measurement of the current flowing through the resistance being measured, when the voltage from the source is constant. To measure resistances in the range from several hundred ohms (Ω) to several megohms (MΩ), the meter and the resistance rx being measured are connected in series. In this case, the current I flowing through the meter and the deflection α of the moving part of the instrument are proportional: I = Cα = U/ro + rx); α = U/C(r0 + rx), where U is the voltage supplied by the source and r0 is the resistance of the meter. For small values of rx (up to several ohms), the meter and rx are connected in parallel. If U and C are constant, the deflection α depends on rx; therefore, the scale of the meter can be calibrated in ohms for easier readout. The error of such an ohmmeter is 5–10 percent of the length of the scale used.

An ohmmeter is frequently part of a multimeter, such as the VOM (volt-ohm-milliammeter). The bridge method of measurement is used in ohmmeters for work requiring great precision. Electronic amplifiers are used to increase the sensitivity of the meter and to improve measurement accuracy.

Electronic ohmmeters with digital readout of the resistance being measured were introduced in the 1960’s. Some ohmmeters can be connected to computers. The range of measurement of such ohmmeters is 1 mΩ to 100 MΩ or more; the error is 0.01–0.05 percent.

### REFERENCES

Shkurin, G. P. Spravochnik po elektro- i elektronnoizmeritel’nym priboram. Moscow, 1972.
Spravochnik po elektroizmeritel’nym priboram. Edited by K. K. Iliunin. Leningrad, 1973.

E. G. BILYK

## ohmmeter

[′ō‚mēd·ər]
(engineering)
An instrument for measuring electric resistance; scale may be graduated in ohms or megohms.
McGraw-Hill Dictionary of Scientific & Technical Terms, 6E, Copyright © 2003 by The McGraw-Hill Companies, Inc.

## Ohmmeter

A portable instrument for measuring relatively low values of electrical resistance. The range of resistance measured is typically from 0.1 microhm to 1999 ohms (&OHgr;).

The ohmmeter solves quickly and easily a variety of measurement problems, including measuring the resistance of cladding and tracks on printed circuit boards, electrical connectors, and switch and relay contacts, as well as determining the quality of ground-conductor continuity and bonding, cables, bus-bar joints, and welded connector tags. See Resistance measurement

McGraw-Hill Concise Encyclopedia of Engineering. © 2002 by The McGraw-Hill Companies, Inc.
References in periodicals archive ?
If the ohmmeter reads less than one megohm, remove the knurled nut that holds the control box cover in place and remove the cover.
They test the system and wiring, using testing devices, such as an ohmmeter and a voltmeter.
Then they find that module in the software and proceed to build a series circuit, determine the series current through each resistor or element, find the voltage across each resister and look at the series through sophisticated simulations of various instruments: an ammeter to measure current, a voltmeter to measure voltage, an ohmmeter to measure resistance and an oscilloscope that lets students see voltage wave forms with instantaneous values.
An ohmmeter is then connected to the power coil and the ground rods.
We planned to give each participant a multimeter--a combination voltmeter, ohmmeter, and ammeter-to use on a dummy circuit board consisting of resistors, LEDs, and test points.
The DLR010HDX 10-A digital low-resistance ohmmeter (DLRO) records tests and has onboard memory storage for up to 200 test records.
The current flowing between the two electrodes is measured in microamperes by an ohmmeter. Measurements above the set norm indicate inflammation or over-activity, and those below the norm indicate blockage or weakness.
Note that the measured resistance of the load is 50.378 [omega] compared to 50.4 [omega], which was measured directly on the resistor using an ohmmeter. Using the Sonnet Suite EM simulator, the knee type of the previously shown structure was simulated.
Resistivity was measured on discs or on the powder between two aluminium plates in an isolated cylinder (diameter = 25mm) using an ohmmeter. In addition, controlled pressure can be applied on the material during the measurement using a screw system.
EAV, also known as electrodermal testing (EDS) or meridian stress analysis (MSA), uses an ohmmeter to measure the electrical resistance and conductivity of specific acupuncture points on the skin.

Site: Follow: Share:
Open / Close