# pitot tube

Also found in: Dictionary, Thesaurus, Medical, Wikipedia.

## pitot tube

[pē′tō ‚tüb]
(engineering)
An instrument that measures the stagnation pressure of a flowing fluid, consisting of an open tube pointing into the fluid and connected to a pressure-indicating device. Also known as impact tube.

## Pitot Tube

an L-shaped tube for measuring the total head of a flowing fluid. The tube is named after the French scientist H. Pitot, who invented it in 1732. The Pitot tube is used as a component of the Pitot-static (Pitot-Prandtl) tube.

## Pitot tube

A device to measure the stagnation pressure due to isentropic deceleration of a flowing fluid. In its original form it was a glass tube bent at 90° and inserted in a stream flow, with its opening pointed upstream. Water rises in the tube a distance, h, above the surface, and if friction losses are negligible, the velocity of the stream, V, is approximately 2gh, where g is the acceleration of gravity. However, there is a significant measurement error if the probe is misaligned at an angle α with respect to the stream. For an open tube, the error is about 5% at α &ap; 10°.

The misalignment error of a pitot tube is greatly reduced if the probe is shielded, as in the Kiel-type probe. The Kiel probe is accurate up to α &ap; 45°.

The modern application is a pitot-static probe, which measures both the stagnation pressure, with a hole in the front, and the static pressure in the moving stream, with holes on the sides. A pressure transducer or manometer records the difference between these two pressures. Pitot-static tubes are generally unshielded and must be carefully aligned with the flow to carry out accurate measurements.

When used with gases, estimate of the stream velocity is only valid for a low-speed or nearly incompressible flow, where the stream velocity is less than about 30% of the speed of sound of the fluid. At higher velocities, estimate of the stream velocity must be replaced with a Bernoulli-type theory, which accounts for gas density and temperature changes. If the gas stream flow is supersonic, or the stream velocity is greater than the speed of sound of the gas, a shock wave forms in front of the probe and the theory must be further corrected by complicated supersonic-flow algebraic relations.

A disadvantage of pitot and pitot-static tubes is that they have substantial dynamic resistance to changing conditions and thus cannot accurately measure unsteady, accelerating, or fluctuating flows. See Flow measurement

## pitot tube

A device, used in conjunction with a suitable manometer or other pressure-reading instrument, for measuring the velocity of air in a duct or water in a pipe.

## pitot tube

An open-end tube that points directly into the air flowing over the aircraft structure, providing pressure for the pitot and, often, the static systems. See pitot pressure.
References in periodicals archive ?
When an internal support frame was present the absolute roughness of the ducts based on Pitot tube test data was 1.
p] is fluid velocity at pass into Pitot tube m/s; [v.
Pitot tube technique is an extensively used experimental method for velocity measurements.
com/prnh/20130307/600769 This report includes the market for altimeter systems, pitot tubes, and other sensors market for the aerospace and defense industry.
Air handling unit supply airflow rate at 'baseline' flow and specific current input values (measured using pitot tube traverses at 'normal operating' fan speed and other points by manually adjusting fan speed at the VFD controller)
AFTER AIRCRAFT OPERATIONS, MAKE SURE YOU TURN OFF THE PITOT TUBE HEATER.
The study also provides market shares for suppliers of primary elements by geographic region, and market shares for suppliers of orifice measuring points, Pitot tubes, Venturi tubes, flow nozzles, and wedge elements.
Tenders are invited for Supply,guarantee and testing of water velocity meter,l-shape pitot tube with differential pressure indicator as pr annexure i ,ii, iii .
According to a new market research report, "Altimeter System and Pitot Tube Market for Aerospace and Defense Industry, Product, by Platform, by Geography (North America, Europe, Asia-Pacific, the Middle-East, Latin America, ROW) - Forecasts & Analysis to 2014 - 2020", published by MarketsandMarkets, the altimeter system and pitot tube market is estimated to be around \$2.
On a third preflight, he found an improperly installed pitot tube.
A malfunctioning Pitot tube could mislead computers controlling the plane to dangerously accelerate or decelerate.
Installs all protective covers, such as engine and pitot tube covers, after the mission is finished.

Site: Follow: Share:
Open / Close