# vernier

Also found in: Dictionary, Thesaurus, Medical, Wikipedia.

## vernier

(vûr`nēr), auxiliary scale, either straight or an arc of a circle, designed to slide along a fixed scale. Its unit divisions, usually smaller than those on the fixed scale, permit a far more precise reading. The vernier is attached to the scales of instruments employed for very accurate linear or angular measurements; these include the transit, sextant, barometer, compass, and caliper. It was devised by a French mathematician, Pierre Vernier, who described it in his*Construction, usage et propriétés du quadrant nouveau de mathématiques*(1631). Certain auxiliary control mechanisms used for fine measurements or adjustments are often called verniers.

## vernier

(**ver**-nee-er) A short scale used to increase the accuracy of the graduated scale to which it is attached. If nine divisions of the vernier correspond to 10 divisions of the instrument scale, the latter can be read to a further decimal place.

## Vernier

**(1)** In instrument making, a device for the exact reading of lengths or angles from the fractional divisions of a scale. The operation of the vernier is based on the eye’s capacity to establish with certainty the coincidence of two lines when one of them is a continuation of the other and their ends overlap. The vernier is a movable scale that can slide along a fixed, basic scale; the divisions on the movable scale are somewhat finer than those on the basic scale. If the interval between the divisions of the basic scale is represented by *a*, and the interval between divisions on the Vernier is (*a - a/n*), then the vernier allows the primary scale to be read with an accuracy equal to *1/n* of its division. The divisions of the vernier are figured in corresponding fractions of the division of the basic scale. If the zero line of the vernier (index) is located between the two lines *c* and *c* + 1 of the basic scale, then the reading is equal to *c* plus that indication of the vernier which is located opposite the line that best coincides with a certain line of the basic scale.

The vernier was invented in 1631 by P. Vernier (1580-1637), the director of the mint in Franche-Comté, and it was named in his honor.

(2) In radio engineering a device for the exact tuning of radio receivers and other radio equipment.

E. A. IUROV

## Vernier

an auxiliary scale used to read fractions of divisions on the main scale of a measuring device. A prototype of the modern vernier was proposed by the French mathematician P. Vernier. [The Russian term, *nonius*, comes from the name of

the Portuguese P. Nunes, latinized to Nonius, who proposed a similar device as a means for reading fractions of scale divisions; this device is no longer used.] There are several types of vernier: linear (Figure 1, a), goniometric (Figure 1, b), spiral (Figure 2), and transverse (Figure 3).

The use of a linear vernier is based on a difference in division intervals between the main scale and the vernier scale. The length of the vernier (a whole number of its divisions) is laid out precisely on an integral number of divisions of the main scale. If the zero mark of the vernier coincides with any mark *L* of the main scale, the measurement result *A* corresponds to the value given by the mark *L*. If the zero mark of the vernier does not coincide with *L*, the value of *A* becomes *A = L + ki*, where

*k* is the number of vernier divisions from the zero division to the division that coincides with the mark on the main scale, and *i* is the smallest fraction of a division of the main scale that can be measured with the vernier (usually *i* = 0.1, 0.05, or 0.02 mm). The principle of reading from a goniometric vernier (used in a

number of opticomechanical instruments) is the same as for a linear vernier. The operation of other types of vernier is explained in Figures 2 and 3.

### REFERENCE

Gorodetskii, I. E.*Osnovy tekhnicheskikh izmerenii v mashinostroenii*. Moscow, 1950.

## vernier

[′vər·nē·ər]## vernier

## vernier

**1.**a small movable scale running parallel to the main graduated scale in certain measuring instruments, such as theodolites, used to obtain a fractional reading of one of the divisions on the main scale

**2.**an auxiliary device for making a fine adjustment to an instrument, usually by means of a fine screw thread